- Bending mechanism & Angulation Control – Good movements
- Light Guide Cable & Light Transmission – Fair
- Objective Lens & Focussing – 8 dark spots on lens and blury optics
- Remote Control Switches – Functional
- Distal End – Fair condition
- Image Quality – Blury image
- Instrument Channel (s) – Clear
- Leak Test – Failed
- Overall Condition – Poor image quality and leaking
- Dimension: 14.8 x 550 mm
For every £100 you spend you will pay £20 Buyers Premium plus £4 VAT on that Premium [non-refundable] and on VAT registered lots £20 Hammer VAT [refundable upon proof of export].
All new and used medical, catering and general equipment lots are sold as seen. Where EST DOM’s (estimated dates of manufacture) are listed these are for guidance only. Any views expressed are for guidance only - please refer to our conditions of sale.
Пн 14 окт 2019 | 10:00:00
Номер лота: 37
Медицинское оборудование | Прямой аукцион
Валюта торгов: £ UK Pound Sterling