- Bending mechanism & Angulation Control – Working
- Light Guide Cable & Light Transmission – No light transmission
- Objective Lens & Focussing – Poor optic, damaged fibres
- Remote Control Switches – N/A
- Distal End – Fair condition
- Image Quality – No image
- Instrument Channel (s) – N/A
- Leak Test – Failed
- Overall Condition – Poor optics and leaking, with scratches on control handle
- Dimension: 4.0 x 250 mm
For every £100 you spend you will pay £20 Buyers Premium plus £4 VAT on that Premium [non-refundable] and on VAT registered lots £20 Hammer VAT [refundable upon proof of export].
All new and used medical, catering and general equipment lots are sold as seen. Where EST DOM’s (estimated dates of manufacture) are listed these are for guidance only. Any views expressed are for guidance only - please refer to our conditions of sale.
Mon 14 Oct 2019 | 10:00:00
Lot Number: 64
Medical | Live Auction
Sale Currency: £ UK Pound Sterling