- Bending mechanism & Angulation Control – Fair movements
- Light Guide Cable & Light Transmission – Good, fibre optic guide cable is not included
- Objective Lens & Focussing – Poor optics several damaged fibres
- Remote Control Switches – N/A
- Distal End – Fair condition
- Image Quality – Poor Image quality
- Instrument Channel (s) – Clear
- Leak Test – Failed
- Overall Condition – Poor image quailty, leaking & damaged sleeve
- Dimension: 5.3 x 400 mm
For every £100 you spend you will pay £20 Buyers Premium plus £4 VAT on that Premium [non-refundable] and on VAT registered lots £20 Hammer VAT [refundable upon proof of export].
All new and used medical, catering and general equipment lots are sold as seen. Where EST DOM’s (estimated dates of manufacture) are listed these are for guidance only. Any views expressed are for guidance only - please refer to our conditions of sale.
lu. 14 oct. 2019 | 10:00:00
Número de Lote: 17
Médico | Subasta en directo
Moneda de Venta: £ UK Pound Sterling