GE Carescape B850 Monitor
Datex-Ohmeda 7900 Smart Vent Ventilator
Datex-Ohmeda Module Rack
GE E-CAIOV-00 Module
Datex-Ohmeda Compact Anaesthesia Monitor
GE E-CAiOV-00 Module
GE E-PRESTN-00 Module
S/5 Compact monitor wirh E-CAiOV gas module and E-PRESTN modules
Central Vacuum Hose
The Carescape Monitor B850 power up and completed the system self-test. The touch screen, C1-CPU, multi gas module and module rack are in good working order. The gas module completed the gas sensor calibration and zeroing and module is detecting and displays CO2 & O2; AA gases were not tested.
Similarly the Compact monitor with both modules working. Temp, SpO2 and NIBP was tested with patient simulator. Good graphic waveforms and parameter values were displayed. The Gas module is working as well. The Compact monitor has SRAM error warning displayed. Lot is in good physical condition with minor signs of wear consistent with use.
For every £100 you spend you will pay £20 Buyers Premium plus £4 VAT on that Premium [non-refundable] and on VAT registered lots £20 Hammer VAT [refundable upon proof of export].
All new and used medical, catering and general equipment lots are sold as seen. Where EST DOM’s (estimated dates of manufacture) are listed these are for guidance only. Any views expressed are for guidance only - please refer to our conditions of sale.
Ends Wed 28 Jul 2021 | 12:30:00
Lot Number: 2
Medical | Timed Online Auction
Sale Currency: £ UK Pound Sterling