- Bending mechanism & Angulation Control – Average movements and locks
- Light Guide Cable & Light Transmission – Fair
- Objective Lens & Focussing – Poor optics
- Remote Control Switches – Working
- Distal End – End cap is not included
- Image Quality – Poor image quality
- Instrument Channel (s) – Clear, forceps elevation mechanism is working
- Leak Test – Failed
- Overall Condition – Poor optics and leaking, control handle with loose parts
- Dimension: 14.0 x 1200 mm
For every £100 you spend you will pay £20 Buyers Premium plus £4 VAT on that Premium [non-refundable] and on VAT registered lots £20 Hammer VAT [refundable upon proof of export].
All new and used medical, catering and general equipment lots are sold as seen. Where EST DOM’s (estimated dates of manufacture) are listed these are for guidance only. Any views expressed are for guidance only - please refer to our conditions of sale.
Mo 14 Okt 2019 | 10:00:00
Medizinbereich | Liveauktion
Verkaufswährung: £ UK Pound Sterling